Product Name£ºPanel Meter 1
Moving Iron Instruments AC Ammeters & AC Voltmeters 
Model: TE96 TE72 TE48 Accuracy: Class 1.5 Specification: 0.1A 0.5A 1A 7.5A 10A 15A 20A 25A 30A 40A 50A 60A 75A 100A For 5A10KA use external current transformer with secondary current 5A and 1A. Notes: For 60A,75A,100A meter. when these meters directly connected to circuit, their accuracy is class 2.5.For 20A60A direct reading TE48 meters, the accuracy is class 2.5. Changeable scale: All scales with printing technique.The dial plate with different scale, which is at the same division and internal value, can be changeable.But the accuracy is unchangeable. Specification: 30V 50V 75V 100V 120V 150V 120V 150V 200V 250V 450V 500V 600VFor 450450KV,meters can use potential transformer with secondary voltage 100V. Heat resistance: Case of noncombustible plastic material. Fastening: By mans of nuts, absolutely safe, fast, robust, sidebyside. Mounting depth: Max.43mm. Heat resistance: Case of noncombustible plastic material. 
Moving Coil Instruments(240DEG,90DEG) DC Ammeters & DC Voltmeters 
Model: TE96 TE72 TE48 Accuracy: Class 1.5 Specification: 0.1A 0.5A 1A 7.5A 10A 15A 20A 25A 30A 40A 50A 60A 75 A100A For 5A10KA use external current transformer with secondary current 5A and1A. Notes: For 60A,75A,100A meter. when these meters directly connected to circuit, their accuracy is class 2.5.For 20A60A direct reading TE48 meters, the accuracy is class 2.5 Changeable scale: All scales with printing technique.The dial plate with different scale, which is at the same division and internal value, can be changeable.But the accuracy is unchangeable. Specification: 30V 50V 75V 100V 120V 150V 120V 150V 200V 250V 450V 500V 600V For 450450KV,meters can use potential transformer with secondary voltage 100V. Heat resistance: Case of noncombustible plastic material. Fastening: By mans of nuts, absolutely safe, fast, robust, sidebyside. Mounting depth: Max.43mm. Heat resistance: Case of noncombustible plastic material. 
Moving Coil Instruments With Rectifier(240DEG,90DEG) AC Ammeters & AC Volmeters 
Model: 90DEG: TEL96 A TEL72 A 240DEG: TEL96A TEL72A TEL48A Accuracy: Class 1.5 Specification: 0.5A 1A 1.5A 3A 5A 7.5A 10A 15A 20A.For10A10KA use external current transformer.Which secondary current is 5A or 1A. Notes: These meters measure r.m.s values and are scaled in r.m.s values. Changeable scale: All scales with printing technique.The dial plate with different scale, which is at the same division and internal value, can be changeable.But the accuracy is unchangeable. Specification: 3V 5V 7.5V 10V 15V 20V 30V 50V 100V 150V 200V 250V 300V 400V 450V 500V 600V.For 450450KV,meters can use potential transformer with secondary voltage 100V. Heat resistance: Case of noncombustible plastic material according to UL94V0. Fastening: By mans of nuts, absolutely safe, fast, robust, sidebyside. Mounting depth: Max.43mm. Heat resistance: Case of noncombustible plastic material. 
Moving Iron Voltmeter With Change Over Switch For AC Ammeters & AC Volmeters 
Model: TEZ96V TEZ72V The instrument are suitable for measuring AC Ampere in three phase 4 or 3 wire networks by using a change over switch on the front side of the instrument. Measuring range: 0.1A / 1A 0.2A / 2A 0.3A / 3A 0.4A / 4A 0.5A / 5A 1A / 10A 5/ 5A 3 phase 3 wire Position: L1,L2 3 phase 4 wire Position: L1,L2,L3 Heat resistance: Case of noncombustible plastic material. The instrument are suitable for measuring AC Ampere in three phase 4 or 3 wire networks by using a change over switch on the front side of the instrument. Technical data: Movement: Movingiron movement with oil filled bearings. practically vibrationfree oil damping. Accuracy: class 1.5 Frequency range: 50Hz. 3 phase 4 wire Measuring range: 60V/100V,80/150V,100/200V,150/250V,250/400V,300/500V. Position: L1L2,L1L3,L2L3,L1N,L2N,L3N 3 phase 4 wire Measuring range: 100V,150V,200V,250V,400V,500V. Position:L1L2,L1L3,L2L3 
Moving Coil Instruments Frequency Meters(240DEG,90DEG) 
Accuracy: Class 1.0 Model: 90DEG:TEF96 TEF72 TEF48 240DEG:TEF96 TEF72 CPF48 Specification: Voltage: 50V 100V 220V 380V 440V 415V Frequency: 4555Hz,4565Hz,5565Hz,4753Hz, 5763Hz,4456Hz,5466Hz,450550Hz,450650Hz,550650Hz Heat resistance: Case of noncombustible plastic material. Fastening: By means of must, absolutely safe,fast,robust,sidebyside. Mounting depth: Max 43mm 
Frequency Meter With Reeds Types 
One row of reeds model: TER96 TER72 Two rows of needs model: TE2R96 Accuracy: Class 0.5 Specification: Voltage: 50V 100V 200V 380V 440V 415V Frequency: 4555Hz,4565Hz,5565Hz,4753Hz, 5763Hz,4456Hz,5466Hz,450550Hz,450650Hz,550650Hz This vibrating reed frequency meter with one row of horizontal reeds or with two rows of horizontally arranged reeds for connection to one main voltage. Vertical arrangements on request Heat resistance: case of noncombustible plastic material. Changeable scale: All scales with printing technique. The dial plate with different scale, which is at the same internal value, can be changeable. But the accuracy is unchangeable. Fastening: By means of nuts, absolutely safe,fast,robust.sidebyside Mounting depth: Max 43mm . 
Moving Coil Instruments Power Factor Meter(240DEG,90DEG) 
Model: 90DEG: TEPF96 TEPF80 TEPF72 TEPF48 240DEG: TEPF96 TEPF72 TEPF48 Accuracy: Class 2.5 Voltage: 100V 220V 380V 415V 440V/current: 1 A or 5A Specification: 0.5cap10.5ind,0.3cap10.1ind,0.4cap10.4ind,0.7cap10.2ind Heat resistance: Case of noncombustible plastic material. Frequency: 50Hz or 60Hz. Fastening: By means of must, absolutely safe,fast,robust,sidebyside. Mounting depth: Max 43mm . 
Moving Coil Instruments Power Meter(240 DEG,90 DEG) 
Wattmetr Model: 90DEG: TEW96 TEW80 TEW72 TEW48 240DEG¡±TEW96 TEW72 TEW48 Varmeter Model: 90DEG: TEVar96 TEVar80 TEVar72 TEVar48 240DEG: TEVar96 TEVar72 TEVar48 Accuracy: Class 1.5 Specification: Voltage: Single phase: 100V 220V 380V Three phase three wire: 100V 220V 380V 415V Three phase four wire: 100V 220(380)V 240(415)V Abovementioned can be connected directly. The following should be connected with voltage transformers:380/100V 500/100V 3K/100V 6K/100V 10K/100V 15K/100V 15K/100V 35K/100V¡ Current: 1A , 5A The voltage circuits of the meter connected with voltage transformers of secondary voltage 100V., if the secondary voltage is lower than 450V,it also can be connected directly. The current circuits connected with current transformers. which secondary current is 5A or 1A .Wattmeters and Varmeters with converter for threephase three wires or four wire and single phase. which can be used for balance and unbalance burder circuit. Heat resistance: Case of noncombustible plastic material. Fastening: By means of must,absolutely safe,fast,robust.sidebyside Mounting depth: Max 43mm. 
Synchroscope Meters 
Synchrony scope for phase comparison for phase AC circuits. 50Hz or 60Hz Model: TES96 TES72 Specification: Voltage: 100V 220V 380V 440V Working time: 15 min max. 
Sequence Indicator 
This indicator is designed for easy testing of phase sequence in threephase system with a permissible operating voltage up to 500V. For short connections, not longer than 5 minutes for one test, a pushbutton switch is placed on the front plate. A disc with an indicating arrow rotates in the direction of the arrow. If the three phases are properly connected,. Otherwise the disc rotates in the other direction. By changing of two wires. The phase sequence will be corrected to corresponding designation of terminals. Model: TESI69 TESI72 Specification: Voltage: 110500V Working time: not longer than 5 minutes. 
Elapsed Time Meter Or Hours Run Meters 
Model: TEH96 TEH72 TEH48 Display: 99999.99 Voltage: AC 10125V AC 200250V AC 380440V AC 24V Burden: 2.5VA Frequency: 50/60Hz Features: Elasped Time Meters or Hours Run Meters monitor ON/RUN time of plant and equipment allowing the user to perform functions such as Production Efficiency Cost Estimating and Service Period Monitorink for preventatives maintenance etc. The time is measured in instrument of 0.01h up to 99999.99 hours after which time the meter resets to zero. Nonreset table to prevent accidental resetting. 
Double Voltmeters 
Model: TE2T96V TE2T72V TE 2L 96V TE 2L 72V Specification: 0100V 0300V 0500V Heat resistance: Case of noncombustible plastic material. Fastening: By means of nuts. absolutely safe,fast,robust,sidebyside. Mounting depth: Max 43mm . 
Moving Iron Instrunment AC Ammmeters For Explosive Gas Atmospheres (Increased Safety "e") 
Model: TEE96 TEE 72A TEE 48A Accuracy: Class 1.5,2.5 Marking: EExe ¢ò T6 Specification: AC 100Ma, 1A , 5A.For 5A 10KA meters have to use external transformer with secondary current 1A , 5A . Frequency: 50/60Hz Consumption: 0.2VA Heat resistance: Case of noncombustible plastic material. Changeable scale: All scales with printing technique. The dial plate with different scale, which is at the same division and internal value, can be changeable. But the accuracy is unchangeable. 
Combined With Moving Iron Ammeters 
For measuring mean maximum content value and instantaneous values for combined with moving iron movement meters. Model : TE 2M 96,TE 2M 72 Accuracy: Class3.0(instantaneous values)Class 1.5 Specification: 8min 5A ,15min 5A ,8min 1A ,15min 1A With bimetal movement and thermal delayed indictor for measuring the maximum mean load during a period of 8 of 15 minutes, eats red maximum pointer. Lead seal can protect resetting knob of maximum pointer. Scale compressed at the beginning. Convenient reading from 25% of fullscale deflection. Accuracy of maximum load indication 3%. Referred to the maximum pointer. Response time 8 or 15,30 minutes(to be specified when ordering). Heat resistance: Case of noncombustible plastic material. Changeable scale: All scales with printing technique. The dial plate with different scale, which is at the same internal value. Can be changeable. But the accuracy is unchangeable. Fastening: By means of nuts, absolutely safe,fast,robust,sidebyside. Mounting depth: Max 43 mm . 
Maximum Demand Ammeter 
For measuring mean maximum current value, when it is 8 min or 15 min or 30 min. Model: TEM96 TEM72 TEM48 Accuracy: Class3.0 Specification: 8min 5A ,15min 5A ,8min 1A ,30min 5A 8min 1A ,15min 1A (No specification for 30 min 5A for TEM48) With bimetal movement and thermal delayed indictor for measuring the maximum mean load during a period of 8 of 15 minutes, eats red maximum pointer. Lead seal can protect resetting knob of maximum pointer. Scale compressed at the beginning. Convenient reading from 25% of fullscale deflection. Accuracy of maximum load indication 3%.. Referred to the maximum pointer. Response time 8 or 15,30 minutes(to be specified when ordering). Heat resistance: Case of noncombustible plastic material. Changeable scale: All scales with printing technique. The dial plate with different scale ,which is at the same internal value. Can be changeable. But the accuracy is unchangeable. Fastening: By means of nuts, absolutely safe,fast,robust,sidebyside. Mounting depth: Max 43 mm. 
Maximum Demand Indicator With Relay 
Model: TEAM96 TEAM72 Features: For the control of Maximum Demand and the monitoring and control of overload on distribution systems. Constructed with an electronic circuit and 5A relay. The alarm was set by the blue slim pointer, the red light is the alarm one, which can be used with buzzer. Accuracy: Class 3.0 Rating: 1A , 5A for use with separate CT. Burden: MD12.2VA Frequency: 50/60Hz Overload: ¡Á 1.2 continuous, 10¡Á1 second,20¡Á1 second via5/5 CT. Relay circuit Auxiliary: AC 220V ¡À10%,50/60Hz Burden:2.6VA Bestrews:<1% Time delay: 1 second approximately fixed. Alarm and Trip point Working Range : 20100% ¡À 1% added to the MDI Accuracy Relay: Single Pole Changeover Relay: Rating: AC 250V 5A non inductive, DC 24V 2A Resistive Temperature: 10 ¡æ to + 50 ¡æ.  
 
